Instant Spice K2 Drug Test Kit
Our state of the art urinalysis K2 drug screen detects the sixteen (16) of the metabolites found in various synthetic cannabinoid products. With this instant and accurate technology, anyone can detect the use of this nationwide abusive drug.
Confirm Biosciences’ K2/Spice on-site test is the most sensitive K2 screen on the market, with the lowest cut-off level 25 ng/ml for synthetic cannabis, compounds used in the majority of K2 products.
Spice and K2 have become the most common slang terms for synthetic marijuana. Others include: Bliss, Black Mamba, Fake Weed, K3, K4 and Yucatan Fire, plus many many more. It is sold under various brand names in head shops and gas stations, often labeled as potpourri or something else. Like any synthetic drug, it can be extremely difficult to tell what is in synthetic marijuana, which is part of the reason it has been linked to numerous dangerous effects and fatalities. However, the Confirm Biosciences’ K2/Spice on-site urine dip strip test kit.
Features and Benefits:
- Up to 72 hour detection
- Results in 5 minutes
- Checks for 16 different compounds
- For Forensic Use
A Total of 16 Compounds Tested
|Name||Compounds Detected by the Confirm Biosciences K2 Test|
|JWH-018||5-pentanoic acid metabolite|
|JWH-073||4-butanoic acid metabolite|
|JWH-210||N-(5-carboxypentyl) metabolite C26H25NO3|
|JWH-398||N-pentanoic acid metabolite|
|MAM2201||N-pentanoic acid metabolite|
|RCS-4||N-(5-carboxypentyl) metabolite C21H21NO4|
Directions to Use:
- Remove the test device from the foil pouch.
- Remove the cap from the test device. Label the device with patient or control identifications.
- Immerse the absorbent tip into the urine sample for 5 seconds. Urine sample should not touch the plastic device.
- Replace the cap over the absorbent tip and lay the device flatly on a non-absorptive clean surface.
- Read results at 5 minutes
Interpretation of Results:
NEGATIVE: Two lines appear. * One color line should be in the control region (C), and another apparent color line adjacent should be in the test region (T).
This negative result indicates that the drug concentration is below the detectable level.
*NOTE: The shade of color in the test line region (T) will vary, but it should be considered negative whenever there is even a faint distinguishable color line.
POSITIVE: One color line appears in the control region (C). No line appears in the test region (T). This positive result indicates that the drug concentration is above the detectable level.
INVALID: Control line fails to appear. Insufficient specimen volume or incorrect procedural techniques are the most likely reasons for control line failure.